Q. Is the mobile theatre compliant with HTM 03-01?
- Yes the ultra clean canopy in the mobile theatre fully complies with HTM 03-01
- All the room air change rates and pressures meet or exceed HTM 03-01
- The air handling unit has heat recovery and is economical to run
Q. What services are required?
- A 400 amp supply is required
- The theatre is 160 KVA
- The recovery unit is 60 KVA
- A 28mm cold water supply is required to each unit
- Local drainage connection
Q. Does the mobile theatre have data connections?
- Yes there is a patch panel within the plant room to data points within the theatre and recovery unit
Q. Does the mobile theatre have a fire alarm?
- Yes there is a Gent stand-alone fire alarm
Q. Does the mobile theatre have a nurse call system?
- Yes there is a Quantec system within the theatre and recovery care
Q. Does the mobile theatre have Med gas?
- Yes there is a manifold for Oxygen, Med air etc.
- There is no NO2 within the theatre
Q. What is the minimum hire period?
- Minimum hire is 6 months
Q. How long does the mobile theatre take to install?
- The mobile takes 1 week to commission once installed on site